Lab Equipment And Lab Supplies | Fisher Scientific
Your source for innovative, indispensable lab equipment, lab supplies, and services. View our wide selection of products for scientific research and education.
Actived: Friday Jan 22, 2021
Instructions - Fisher Sci
the electrode and can be simply be rinsed off with clean water. To remove the storage bottle from the electrode, first unscrew the lid completely then separate from the bottle—pulling the electrode directly out of the bottle before loosening the cap may damage the junction. Please remove the storage bottle from the end of the electrode before
Instruction Manual CON 6/TDS 6
The CON 6 / TDS 6 meter has 6 keys on its splash-proof keypad; ON/OFF, HOLD/ENTER, CAL, MODE, and keys. Some buttons have several functions depending on its mode of operation. ON OFF • Powers on and shuts off the meter. Takes you directly into measurement mode when meter is switched on. CAL
CON 11 & CON 110
OFF ON • ON/OFF - Powers on and shuts off the meter. The meter will start in the measurement mode it was in when last switched off. MEAS CAL • CAL - Activates the Conductivity or TDS calibration mode and when used with the MODE key, it activates the temperature calibration mode. • MEAS - Allows return to measurement mode when canceling or
Large Screen Waterproof ORP / Redox Tester
Press ON/OFF to turn off tester. If you do not press a button for 8.5 minutes, the tester will automatically shut off to conserve batteries. User Reset You can reset the ORP calibration to the factory default by using the user reset function. 1. Switch off the tester. 2. While pressing the ‘Cal’ button, press and release the ON/OFF button
accuTupH® / accuTupH+® Rugged Bulb Combination pH Electrodes
the electrode and can be simply be rinsed off with clean water. To remove the storage bottle from the electrode, first unscrew the lid completely then separate from the bottle—pulling the electrode directly out of the bottle before loosening the cap may damage the junction. Please remove the soaker bottle from the end of the electrode before use.
Instruction Manual: Portable Colorimeter
The keypad has four keys: ON/OFF, MODE, ZERO and READ/ENTER. Keys Functions ON/OFF - Powers on and shuts off the meter. With the auto switch-off feature the meter automatically shuts off 10 minutes after last key press. MODE - Selects the test to be performed. - Selects the calibration point in an incremental circular manner during calibration.
RTD THERMOMETER MODEL NO. 60010-85
1. Before changing battery, turn instrument off and disconnect probe. 2. Loosen screw and lift battery cover off back of case. 3. Remove the two AA batteries. 4. Insert two new batteries, observing polarity. 5. Install cover and tighten screw. AC ADAPTER. An optional AC adapter can be used for power to conserve batteries. The adapter is not a
Instruction Manual: Oakton pHTestr 10, 20, 30, 10BNC, Spear
off, press CAL and ON/OFF at the same time. First release ON/OFF button, then CAL button. After reset, whole LCD segment lits up and goes back to measurement mode. Figure 4: Example of HOLD Function 3. Press the CAL button to enter calibration mode. The ‘CAL’ indicator will be shown. The upper display will show the measured reading based on